ARCHITECTURAL SPECIFICATIONS & ROUGH-IN INFORMATION
STANDARD TUB - MODEL MB-80
CONSOLE TUB - MODEL MB-80R
|Width||Standard (MB-80)||34 ½"|
|Width||Console (MB-80-R)||52 ½"|
|Tub Rim Height||(All Models)||36 ½"|
|Console Height||Console (MB-80-R)||68"|
|Back Flow Prevention:||1" Air gap for tub fill, vacuum breakers for MB-80-R shower hoses. Watts N-9 double checks for MB-80 shower hoses. Note: The factory provided backflow prevention is the most widely accepted. Local codes may require changes, which is the resposibility of the local plumbing contractor.|
|BathAire(SM) Outlets:||5 with 6 ports each for a total of 30 warm air massage outlets.|
|Door Seal:||Seal certified by Warnock Hersey to the ASME A112.19.15-2005, and ASTM D 2000 standards to ensure long lasting durability for bathtubs with pressure sealed doors. Highly resistant to oils, detergents and all water conditions.|
|Electrical:||UL/CSA approved 120V AC, 60 Hz, 15 Amp dead front GFCI (ground fault circuit interrupter)|
|Installation Time:||1 ½ hours|
|Inspection and Adjustments:||½ hours|
|Liquids Storage||Located behind lockable service access doors.|
|Motor Specs:||BathAire(SM) Motor: 120 V, 11 Amp, 1.3 H.P. Variable speed with built-in 600 watt heating element for maximum comfort and control.|
|Patent:||US 6,7666,543 B1.|
|Room Size:||Standard Model 6' wide x 9'4" long: Console model 6' wide x 10' long. Can fit in smaller room; call MasterCare with dimensions.|
|Services Access||Controlled access through lockable service doors.|
|Electrical Standards:||CAN/CSA C22.2 No. 218.2-93 (R2006) Hydromassage Bathtub Appliances.
UL 1795 Hydromassage Bathtubs, Third Edition, Including revisions through to Sept. 22, 2006.
|Plumbing Standards:||Warnock Hersey: CSA B45.5 (2008), ASME A112.19.15 (2005), CSA B45S1 (2004), ANSI Z124.1.2 (2005).|
|State Standards:||IL Dept. of Public Health, Product approval letter dated 10/28/04.
Commonwealth of Massachusetts, Approval Code P3-1108-235
Michigan State Code Commission, Certificate of Acceptability No. 1554 PA.
Wisconsin Administrative Code. In Compliance with chapters Comm. 82 through 84, Wis. Adm. Code, and Chapters 145-160, Wisconsin Statues. Product File #20060062 & 20010143.
|Thermometer:||Digital temperature measurements.|
|Tub Entry:||Door counter-balanced with simple guidance system for ease of operation.|
|Tub Fill/Shower:||On-board high volume maxim valve tested to ASSE-1016-2005 and volume controlled shower wand with back-flow prevention tested to ASME A112.18.3|
|Tub Shell:||Sanitary gel-coated FRP (Fiberglass Reinforced Polymer).|
|Tub Water Use:||55 gallons minimum with normal displacement 4 inches below overflow escutcheon.|
|Waste:||2" PVC schedule 40 pipe.|
|Water Connection:||¾” hose swivel connections supplied by manufacturer.|
|Water Supply:||¾” Copper hot and cold. Minimum of 30 PSI for hot and cold required. Maximum temperature of hot water supply 115°F or code, whichever is less. Shut off valves at unit supplied by local plumbing contractor.|
|Weight:||MB-80 (420 lbs), MB-80-R (Tub 385 lbs, Console 240 lbs), MT-80 Transfer system (150 lbs).|
|Useful Life:||Approximately 10 years -- determined by use and conditions.|